To the configurator
Thermocouples 1(2) x Type “J”, “K” and “N”, Basic values according to DIN EN 60584, tolerance class 1 or 2; Type “L” according to DIN 43710, tolerance class 2; installed in a thermal absorption box with a diameter of 6x6x30 mm made of copper, brass or VA, connected compensation cable with cross-section 2×0.22 qmm, individual teflon and together with silicone, cable lengths can be selected individually. A worm drive clamp is fixed to the box for fixing on the pipe. The receiver box is made along the length with the required radius of the piping, e.g. radius R12; corresponding to the additional type specification (x) for special feature.
(For variants please refer to the worksheet).